1. Fill the sample in sample cup until it reaches the level line given inside. 2. open the glass draught screen and Pivot the DIPs Pod upwards. 3. Place the lid on the sample cup and it will sit perfectly in one orientation. 4. Now insert the cup and lid into the test area, the cup exactly sit in one position. 5. Adjust and make sure the igniter should be like this given. 6. the DIPs Pod down until it latches in position, engaging the stirrer drive shaft into the stirrer. 7. Make sure the shutter actuator on the DIPs between the shutter drive pins. 8. Close the glass draught screen and Power ON the device, wait for the start up. 9. Main menu will appear on screen, move the cursor using Alter Parameters by and select pressing 10. Change the Expect Flash Point value, if needed. Press, hold and turn the knob Note: if the sample temp too high or bath temp too high message has appeared on line, change the expect flash point value to high 11. Press and it will goes to the main menu then select Run Test. and 12. Edit sample name using'O +O and press next button 2times. 13. When the test starts the display will change and shows the status. 14. To abort the test at anytime pressing the stop button. 15. After the test complete result will be displayed, turn to see completely 16. To see the result after the test, go to main menu select the show result by pressing 2 times. WARNING: the test area may be very hot, with temperature up to 400°C.